VNS Therapy® Magnet FAQs
Find answers to commonly askedddddd questions about the VNS Therapy® magnet in our knowledge base
The magnet is used to provide an extra dose of therapy if a breakthrough seizure occurs and to temporarily suspend therapy.
The VNS Therapy® generator has a sensor called the Reed switch. The Reed Switch can be activated and deactivated by the magnet which will start or stop stimulation.
Use the magnet for extra stimulation as often as you like, but for no more than 8 hours (continuous stimulation). Constant or frequent use of the magnet will use up the battery in the generator and could hurt the nerve. If you use the magnet often, your device settings may need to be adjusted. This is something to discuss with your doctor.
Depending on the settings, using the magnet will provide an additional 7- 60 seconds of stimulation each time you use it. Using the magnet again during the same period will not make the stimulation stronger but will restart the magnet stimulation. Wait until the stimulation ends before trying it again.
You should receive two magnets when your device is implanted. Please contact your doctor for additional magnets.
Results from magnet stimulation differ for each person.
The magnet should be kept at least 10 inches (25 cm) away from credit cards, televisions, computers, microwave ovens, and other magnets.
No, the VNS Therapy® Magnet is the only magnet that should be used with the VNS Therapy System.
No. VNS Therapy® automatically delivers therapy at regular intervals throughout the day in an effort to control seizures. The magnet is an optional benefit that may provide additional seizure control should a breakthrough seizure occur.
With normal use, the VNS Therapy® Magnet should last about 3 years.
Yes. To stop stimulation, hold the magnet over the generator and keep it there (some people use tape). If you have unusual or painful stimulation, contact your doctor immediately.
Use the magnet under the following circumstances:
- To start stimulation when you feel a seizure coming on or any time during a seizure
- To stop stimulation if it is painful or you plan to give a speech or sing
- To test that the device is operating properly
The magnet’s strength should not be affected if it is dropped. This is a common problem with low-power magnets.
The magnet overrides your normal treatment schedule, whether or not the device is ON at that time. Once the magnet-activated stimulation ends, the device will return to the treatment schedule set by your doctor.